In all departments of the hospital, the use of medical endoscope is very strict, including the use of its components. The endoscope equipment maintenance company will explain the relevant contents for you.
Indoor instrument positioning:
The power socket in the operating room should be roof-mounted to avoid the interference of the power cord caused by the staff moving and touching the power socket, so as to ensure the smooth operation and avoid the damage of the instrument due to the unstable voltage.
Generally, the monitor has been tested and adjusted to a good display state by relevant technical personnel during installation, so it is not recommended that users debug by themselves. If the monitor does not display, check the connection of the patch panel and the power cord, whether the display switch is on, whether the camera system is on, and whether the output of the camera system is correctly connected to the input port of the monitor.
After color distortion, the optical interface of the camera system can be aimed at objects or pictures with bright colors, and the contrast, saturation and brightness can be adjusted through the menu on the monitor. Avoid strong magnetic objects close to the monitor, and perform degaussing maintenance regularly. In case of serious magnetization or degaussing failure, please contact the manufacturer's technical personnel.
Camera system and light source:
Although many medical endoscopes declare that the optical interface is waterproof, it is not recommended to disinfect the optical interface and the connecting line. It can be put on the germ-proof sleeve and sterilize the germ-proof sleeve every time it is used, which can greatly increase the service life of the optical interface. If it is necessary to disinfect, please use the method of low temperature fumigation.
The socket cannot be forcefully inserted and pulled out. It should be directly inserted and pulled out at the position of 12 o'clock, handled gently, and regularly calibrated for white balance. Prevent the optical interface from contacting with liquid, even if it is claimed to be waterproof equipment, so do not immerse and disinfect the focusing head and wiring. If it does not contact with liquid such as glucose, please wipe the liquid on the surface of the optical interface with a dry cloth immediately after the operation. Prevent inflexible movement of bayonet crystallization. If there is any abnormality, please do not force to use it.
The optical cable cannot be disinfected by immersion or high temperature and high pressure. If it needs to be disinfected, please use low temperature fumigation or light alcohol to wipe the surface. The user should carefully record the service time of the light source bulb and prepare the spare bulb when it is close to its service life. The light source bulb has a limited service life. It is recommended to turn off the light source immediately when it is not in use.
The requirements for the use environment of the endoscope are that the indoor environment is mainly darkroom environment, avoiding direct sunlight, turning off the fluorescent lamp at the beginning of the operation, using shadowless lamp for lighting, and maintaining the indoor temperature at 22 ℃~25 ℃, humidity at about 50%, acid, alkali and vapor proof. Avoid interference of strong current and electromagnetic field. Come to our website for more matters Consult and understand!